Your Top Choice for Scales!
Call: (919) 776-7737
The Transcell TI-500E is a general purpose, "no frills" certified digital indicator housed in a durable black abs enclosure. The official item number is P104190372 but we typically just call it the TI500E indicator. It comes with a large (0.6") green LED screen for easy read out & supplies enough current for up to four 350-ohm load cells. THIS IS A GREAT REPLACEMENT DIGITAL READOUT for an old scale. Economical, easy to setup, virtually trouble free use.
All TI 500E setup parameters and calibration parameters may be entered via the front panel keys. The TI-500 Digital Weight Indicator is a simple device and uses full duplex RS-232 serial format for communication with many types of attached peripheral devices. The unit can transmit data on demand, or continuously in a popular data protocol to match a wide variety of printers, remote displays, or personal computers.
Mechanical Conversion Kit
Floor / Deck Scale Indicator
TI Series Indicators Standard Features
High Quality / Low Cost
NTEP Approved for 5000 divisions
Full Duplex RS-232 Serial Port
Easy to Read LED Display
Displays up to 50,000 graduations
Drives up to four 350-ohm Load Cells
Full Front Panel Configuration
Stainless Steel Swivel Stand
TI 500 E Digital Weight Indicator features
ABS NEMA 12 enclosure
NTEP approved for 5000 divisions
Gross, net, tare, print, ticket mode
Full front panel configuration
Indicator Dimensions: 3.2" x 6.8" x 2.3"
Stainless steel swivel stand
115VAC power / AC adapter included
TI-500E comes w/ 15' interface cable (load cell cable)
The Transcell TI500e is a relatively simple device and is almost a "plug and play" solution. However, you will need to do a few things in order to make this digital weight indicator work with your existing scale.
Please read the scale manual. When you receive this unit, you will need to connect the included load cell cable to your existing summing board. The standard color code of the cable is Red + Exc, Blk - Exc, Grn +Sig, White -Sig.
Then, you will need to setup and configure the indicator using the settings inside the TI series indicator (primarily F1, F9, & F10). Then, you will need to do a zero calibration (F16) and a span calibration (F17). For the span cal, you will need to use a known weight that is ideally over 10% of the capacity. You can read more in depth instructions on the calibration by looking at page 10 of the manual.
If you follow the instructions in the TI 500E manual carefully, this indicator is one of the easiest weight indicators to setup & calibrate on the market.